Files in this item
Can automated imaging for optic disc and retinal nerve fiber layer analysis aid glaucoma detection?
|dc.contributor.author||Burr, Jennifer M.|
|dc.identifier.citation||Banister , K , Boachie , C , Bourne , R , Cook , J , Burr , J M , Ramsay , C , Garway-Heath , D , Gray , J , McMeekin , P , Hernández , R & Azuara-Blanco , A 2016 , ' Can automated imaging for optic disc and retinal nerve fiber layer analysis aid glaucoma detection? ' , Ophthalmology , vol. 123 , no. 5 , pp. 930-938 . https://doi.org/10.1016/j.ophtha.2016.01.041||en|
|dc.identifier.other||PURE UUID: b536ef37-2bbe-42f5-bc44-2c832b894e94|
|dc.description||Open Access funded by Department of Health UK||en|
|dc.description.abstract||Purpose: To compare the diagnostic performance of automated imaging for glaucoma. Design: Prospective, direct comparison study. Participants: Adults with suspected glaucoma or ocular hypertension referred to hospital eye services in the United Kingdom. Methods: We evaluated 4 automated imaging test algorithms: the Heidelberg Retinal Tomography (HRT; Heidelberg Engineering, Heidelberg, Germany) glaucoma probability score (GPS), the HRT Moorfields regression analysis (MRA), scanning laser polarimetry (GDx enhanced corneal compensation; Glaucoma Diagnostics (GDx), Carl Zeiss Meditec, Dublin, CA) nerve fiber indicator (NFI), and Spectralis optical coherence tomography (OCT; Heidelberg Engineering) retinal nerve fiber layer (RNFL) classification. We defined abnormal tests as an automated classification of outside normal limits for HRT and OCT or NFI ≥ 56 (GDx). We conducted a sensitivity analysis, using borderline abnormal image classifications. The reference standard was clinical diagnosis by a masked glaucoma expert including standardized clinical assessment and automated perimetry. We analyzed 1 eye per patient (the one with more advanced disease). We also evaluated the performance according to severity and using a combination of 2 technologies. Main Outcome Measures: Sensitivity and specificity, likelihood ratios, diagnostic, odds ratio, and proportion of indeterminate tests. Results: We recruited 955 participants, and 943 were included in the analysis. The average age was 60.5 years (standard deviation, 13.8 years); 51.1% were women. Glaucoma was diagnosed in at least 1 eye in 16.8%; 32% of participants had no glaucoma-related findings. The HRT MRA had the highest sensitivity (87.0%; 95% confidence interval [CI], 80.2%–92.1%), but lowest specificity (63.9%; 95% CI, 60.2%–67.4%); GDx had the lowest sensitivity (35.1%; 95% CI, 27.0%–43.8%), but the highest specificity (97.2%; 95% CI, 95.6%–98.3%). The HRT GPS sensitivity was 81.5% (95% CI, 73.9%–87.6%), and specificity was 67.7% (95% CI, 64.2%–71.2%); OCT sensitivity was 76.9% (95% CI, 69.2%–83.4%), and specificity was 78.5% (95% CI, 75.4%–81.4%). Including only eyes with severe glaucoma, sensitivity increased: HRT MRA, HRT GPS, and OCT would miss 5% of eyes, and GDx would miss 21% of eyes. A combination of 2 different tests did not improve the accuracy substantially. Conclusions: Automated imaging technologies can aid clinicians in diagnosing glaucoma, but may not replace current strategies because they can miss some cases of severe glaucoma.|
|dc.rights||Crown Copyright 2016 Published by ELSEVIER Inc. on behalf of American Academy of Ophthalmology This is an open access article under the Open Government Licence (OGL) ( http://www.nationalarchives.gov.uk/doc/open-government-licence/version/3/ )||en|
|dc.subject||RA0421 Public health. Hygiene. Preventive Medicine||en|
|dc.title||Can automated imaging for optic disc and retinal nerve fiber layer analysis aid glaucoma detection?||en|
|dc.contributor.institution||University of St Andrews.School of Medicine||en|
This item appears in the following Collection(s)
Items in the St Andrews Research Repository are protected by copyright, with all rights reserved, unless otherwise indicated.